Seta Reid Vapour Pressure Bath
Part Number 22210-4
The Seta Reid Vapour Pressure Bath is a floor standing water bath specifically designed for the determination of vapour pressure with Seta Reid or Seta Vapour pressure cylinders.
The stainless steel 80 litre water bath has a temperature range of ambient +5°C to 90°C and can achieve a test temperature of 37.8°C (100°F) with a uniformity of ±0.1°C (0.2°F).
A programmable, digitally controlled thermostir unit combines the functions of heater, temperature controller, timer and stirrer. The thermostir unit is equipped with an RS 232 port, a switched output for an external relay, a socket for an external PT100 temperature probe and alarms and cut-outs for low fluid level and over-temperature.
The 22210-4 will accept up to four 22410-0 and up to six 22400-0 Seta Vapour Pressure Cylinder / Ullage Chamber assemblies.
Supplied with: 4 pressure cylinder mounting brackets, mains lead and instruction manual.
|Test Temperature Range||ambient +5°C to 90°C (9°F below ambient to 194°F)|
|Temperature Uniformity||± 0.1°C @ 37.8°C|
|Bath Volume||80 litre|
|Bath Fluid||Water (De-ionised)|
|Cylinder Capacity||Up to 6 cylinder assemblies|
|Size (HxWxD)||77 x 55 x 51cm|
|CCCN Code:||Tariff 84198998|
DETAILS OF METHODS
|ASTM D1267||Standard test method for gauge vapour pressure of liquefied petroleum (LP) gases (LP - gas method)|
|IP 410||Liquefied petroleum gases - Determination of gauge vapour pressure - LPG method (ISO 4256:1996)|
|ISO 4256||Vapour Pressure of Liquefied Petroleum Gases|
|ASTM D4953 Procedure A||Standard Test Method for Vapor Pressure of Gasoline and Gasoline-Oxygenate Blends (Dry Method)|
|IP 402 (obs)||Vapour Pressure of Petroleum Products (Reid Method)|
|ISO 3007||Vapour Pressure of Petroleum Products (Reid Method)|
|EN 12 (obs)||Vapour Pressure of Petroleum Products (Reid Method)|
|FTM 791 1201||Vapour Pressure of Petroleum Products (Reid Method)|
|BIS IS 1448 Part 39||Vapour Pressure by Reid Method|
|BS 2000 Parts 69, 402 (obs) & 410||Petroleum products - Determination of vapour pressure - Reid method|
|ASTM D323 Procedures A, C & D||Standard test method for vapor pressure of petroleum products (Reid Method)|
|IP 69 Procedures A, C & D||Petroleum products - Determination of vapour pressure - Reid method|